This product is a analog output which can be coupled to the ToolScope via EtherCAT. You need a bus coupler E65 19010 for that.
Short description of the delivered product (Quotation text)
Additional information to the delivered product
The connection is done by an Ethernet cable via the EtherCAT protocol.
The ToolScope is connected to a EtherCAT bus coupler. It communicates with the coupler. The modular input and output units are added to the coupler.
BRINKHAUS delivers as standard unit
To supply the above mentioned modules with power and to connect them via EtherCAT to the ToolScope, one needs a bus coupler. This is E65 19010.
|Task in the EtherCAT system||Deliver analog output values|
|Count of terminals||4 (2 wires, single ended)|
|Load impedance||> 5 kΩ (short circuit protected)|
|Power supply||24 V DC|
|Conversion time||~ 100 µs|
|Output resolution||16 Bit (incl. sign)|
|Width in process image||8 bytes|
|Error in output||<0.1% of nominal range|
|Electric isolation||500V (Power contact / Power supply / Ethernet)|
|Current consumption E-Bus||140 mA typ.|
|Certificates||CE, UL, Ex|
|Mechanic shock resistance||EN 60068-2-6/EN 60068-2-27|
|EMC resistance||EN 61000-6-2/EN 61000-6-4|
|Ambient temperature||-25…+60 °C/-40…+85 °C (95% humidity, non condensating)|